Your basket is empty...
The linear velocity is an important parameter in chromatography. It has an influence on chromatography resolution and therefore setting the dead volume is basic part of method development.
To set a dead time, inject 2 µl of a non-retained gaseous substance, which is compatible with the detector. Take a gas-tight syringe and draw the headspace over neat compound. Accurately mark the injection starting time and peak elution time.
|FID||methane, propane, butane|
|ECD||methylen chloride, air (at lower oven temperatures)|
|TCD||methane, butane, air (at lower oven temperatures)|
|NPD||acetonitrile, air (at lower oven temperatures)|
|MS||propane, butane, argon, air (at lower oven temperatures)|
Note: Some compounds may be slightly retained on thick-film phase, however, they will be reproducible for similar column types.