For checking and adjusting procedure folow instruction of SP01. Is necessary to remove both sides, front and top panel (4mm allen key). Be carreful when you remove top panel because of flat cable between display and motherboard. This panel can be leaning on the right side.
If you want to check 4 fans and order of turning-on (turning-off) of compressor you have to remove right metal cover. The back compresor (312) has to start before front compressor (332) and stop after front compressor (332. Note:Refit metal cover before next tests because of proper cooling.
Use “LNI-Schmidlin_Internal info and configure form” for checking and adjustment of all values
Check output pressure with on and off output valve (normal diference is 0.5 to 0.6 bar)
Check ouptut flow
Check oxygen sensor flow
Check all gauges on left side. Note: Pump pressure during cycle switching can’t exceed of gauge range (adjustment by relief valve near coalescing filter – two allen nuts for lock of position)
Check and adjust both regeneration flow during corresponding cycle with opened output valve.
Check oxygen level by external detector .
Check all other items.
Suggestions and notes:
Ouput presure must be stable during both cycles. Check this pressure with closed and opened output valve (normal diference is 0.5 ÷ 0.6 bar).
If its necessary to adjust ouput pressure, indication of correct adjustment of Section 1 is good sensitivity of pressure change to knob rotation.
In the case some dysfuncion (wrong pressure or leakage) check gauges and solenoid valves by external gauge which is connected to the adequate point.
Suction filters may serve as silencers on the outputs.
Exhaust of nitrogen excess is near to oxygen sensor (some models are equipped with a silencer)
Water from compressed air drains to the inside hot part of generator and vaporizes.
Higher pressure in the Line 1 and Line 2 usually has an impact on nitrogen purity
Adjustment of section 2 is very important (with open output valve). External gauge is connected instead of built-in (front bottom right). Pressure is usually 2 bar (maximum is 2.2) and absolutely stable on built/in gauge. When is opening the needle valve, pointer on the gauge starts to vibrate, so little bit close the needle valve (for stop of vibration) and then check output opressure and flow.
Very often is in the SP’s wrong description of this gauge (is stated Section 1 instead of Section 2). However postion of this point is between Tank and second head of front compressor (332).
Laurent very often changed in configuration Capacity time to 9 cec and Volume to 6.1 L/min (influence to flow calculation).
Digital pressure sensor S#1 (visible in diagnostic) measures pressure after coalescing filter (before Molsieves) and pressure changes during each cycle.